Manometers MP KIMO MP -K: from to °C -J: from to °C -T: from to °C CO/Temperature: CO: from 0 to ppm, from Consult KIMO’s MP brochure on DirectIndustry. Page: 1/4. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air.
|Published (Last):||15 April 2017|
|PDF File Size:||7.27 Mb|
|ePub File Size:||10.66 Mb|
|Price:||Free* [*Free Regsitration Required]|
On disabled mode, the probe is connected, the measurement is not carried out and the value is not displayed. When CO peak is selected, the period is diplayed 30 seconds by default.
Kimo MP200 Thermo-Anemo-Manometer
Air velocity with Pitot tube: Launch a planned dataset A planned dataset is composed of several locations. Select compensation It is possible to modify the value of the compensation in temperature. Average – Rec Recording Veloci. Click on Esc to return to previous screen. Click on OK to validate.
Battery When battery indicator flashes it is recommended to change the batteries: Press 1x in order to select HOLD function: To save jp dataset click on Save with the access key. With Pressure function, you can access to following sub-functions – Hold – see Air velocity – Config.
If the probe is connected for more than 1 minute, measurement screen appears and indicates the concentration and places it on a bargraph. This sub-function allows to compensate for any long-term drifts of the sensing element by a manual adjustment of the zero.
When the solenoid valve is enabled, it runs every minute. This menu allows to view probe information plugged to C2, Module, C1 or wireless connections. For each location, the operator can enter a theorical value and a tolerance for the parameter to be controlled. Create or launch a continuous dataset Select dataset A continuous dataset can be carried out using MP and is composed of several dated measuring points.
Numbers of selected points and parameter for which calculation is carried out, are displayed For adding a new measuring point to this calculation, press OK to confirm. Select Start with access key for launching measurement. If K factor is selected, you must enter value. Click on Mesure to display values table of selected dataset. Memory capacity of the instrument is up to 8, points or 50 datasets. Select Start for dataset launching.
From this screen you can: Probe display Available information are: The numbers of points is displayed.
You can choose between a planned or a continuous dataset. Configuration – Params Parameters – Avg. Probe Hold Alarms Params Veloci.
The number of points selected and the parameter are displayed. You can access this function selecting COmax with the access key CO max The CO is measured on an adjustable period, the maximum value measured in this period is called CO max. Use the arrows to modifiy the threshold value then press Validate to validate alarm threshold and back to measurement screen. Configuration sub-function allows to: When the pm is finished, the CO max is displayed.
Kimo MP H MPH Thermo-Anemo-Manometer
Planification must be made via the software. Configuration If you use thermocouple probes, you must enter type into the Configuration sub-function.
Press Ok for launching measurement. The ”Probe” menu only appears when probes or module are connected. When the measurement is fixed, press Min Max to access to maximum and minimum values of kiko measurement.
The beep repetition frequency will increase accordong to the increase of detected gas concentration, and the display will blink at the same frequency. From kmo screen, you can: If you click on average icon, max. Enter dataset time and interval of time between 2 measurements by selecting Period with access key. Repeat this operation for the month, year, hour and minute. Wire probes with Smart-plus system Wire probes with Smart-plus system are automatically recognized when connected to the instrument.
MP Portable Thermo-Anemo-Manometer
If Other is selected, you must enter a value. Preview of tables of points of datasets Dataset summary You can display tables of points of datasets performed on the device. It’s a community-based project which helps to repair anything.
Go to Recording menu.
Remove the front part at the back of the instrument. This datasets can’t be set using Datalogger Software. Once kiml Autoz key, the zero is readjusted.
If the probe has been just connected, the screen indicates the remaining time of warm up then the measurement screen appears, indicates the concentration and places it on a bargraph.